CSA C22.2 No. 61010-2-061:19
Safety requirements for electrical equipment for measurement, control, and laboratory use — Part 2-061: Particular requirements for laboratory atomic spectrometers with thermal atomization and ionization
1 Scope and object
This clause of Part 1 is applicable except as follows:
1.1.1 Equipment included in scope
Replace the text, except the first paragraph, with the following new text:
This part of IEC 61010 applies to electrically powered laboratory atomic spectrometers with
NOTE 1 Examples include atomic absorption spectrometers, emission flame photometers, atomic fluorescence
spectrophotometers, inductively coupled plasma spectrometers, microwave coupled plasma spectrometers and
mass spectrometers, all with thermal atomization and ionization (including tubing and connectors which are
provided by the manufacturer for connection to external supplies).
NOTE 2 If all or part of the equipment falls within the scope of one or more other Part 2 documents of IEC 61010
as well as within the scope of this document, consideration is given to those other Part 2 documents.
1.1.2 Equipment excluded from scope
Add, before the first paragraph, the following new text:
This document does not apply to thermal atomization detectors (flame ionization detectors)
used in gas chromatography.